How to 3D Print Multicolored Models with Formlabs SLA
Mr. Shehryar Khan has 5 years of hands-on experience with 3D Printing technologies and a strong background in research, developing and preparing new applications for the 3d printing medical marketplace.
Materialise DLP Competence Center Malaysia is focused on DLP 3D Printing technology. It has the ambition to investigate novel uses of the DLP technology and tailor Materialise software technology to maximize its use. The Materialise DLP Competence Center is staffed with highly specialized engineers and equipped with a state-of-the-art research lab.
Image-based medical software and 3D Printing have revolutionized the health industry. Health experts across the globe can, for example, create tangible 3D-printed replicas of the patient’s anatomy. Mr. Khan, what’s the added value of multicolored models?
“A deep understanding of the pathological anatomy is key to planning a treatment procedure that aims to achieve the best patient outcome possible. Printing in one color is useful if you need to visualize just one type of tissue – such as complex trauma fractures, a CMF reconstruction or visualization of a heart disease. Multicolored models, by contrast, become particularly relevant when you need a way to differentiate between tissues; such as tumor and surrounding organ tissue, or veins and nerves in an organ.”
But most desktop 3D printers in hospitals only offer a single color solution.
“Correct. The only available 3D printing technology enabling multicolor 3D Printing would be the Multi Jet Printing and Binder Jetting. We do notice more and more hospitals invest in implementing 3D Printing in their workflow – especially after achieving the significant milestone of receiving an FDA clearance for the Materialise Mimics inPrint software in junction with a desktop 3D printer for medical use to print patient-specific anatomical models or diagnostic purposes. Implementing an industrial 3D printer that enables color printing would require a major investment on the part of the hospital in terms of finance, space, expertise, and so on. This is why we wanted to explore the possibility of creating a multicolored 3D-printed anatomical model using a desktop 3D printer (Formlabs SLA) and the Materialise Mimics inPrint software.”
Desktop 3D printers are resin-based. What resin is recommended to create a multicolored model?
“To ensure the best outcome, using transparent resin to execute the actual 3D printing is recommended. Once printed, it is important to check for resin residue trapped in small chambers or cavities. Air spray can be very helpful at this point! After the internals are cleaned, the next step is to remove the supports, smoothen the surface and dry the part.”
We now have a transparent 3D-printed model, but how do we add color to it?
“You can paint your 3D-printed anatomical model by injecting color into the sections through the drain holes. To do so, you create a balanced mixture of water, acrylic paint and primer: this is essential to achieve a perfect color adhesion. The paint-to-primer-to-water ratio should be adapted according to the painting tools you are attempting to use, but the recommended paint-to-primer-to-water ratio would be: for syringe [needle specs] injections - 1:1:1 or 1:1:2; For pipette [outlet specs] injections: 2:2:1. Once dried, the colored part undergoes surface finishing through sanding, glossing and UV curing.”
Creating a multicolored 3D-printed model using medical software and a desktop 3D printer seems like a cost-efficient alternative for the expensive industrial options.
“The workflow I’ve just described allows for the creation of multicolored 3D-printed anatomical model in hospital 3D printing labs destined for, in particular, research, communication, and educational purposes. This can indeed be done at a relatively low cost, but at the expense of some manual post-processing work. That being said, this 3D printing workflow will result in some limitations once you start designing fine anatomies. For example: since a wall thickness of 1 mm is used for hollow sections, parts that are smaller than 2mm in size won’t be able to be hollowed. In fact, it appears that the internal diameter of the hollowed section needs to be greater than 1 mm for the hollow to print. Hence, small parts cannot be hollowed and 3D-printed. Additionally, small hollows are more susceptible to air bubbles.”